RSP-F35 RF & LED Mask Machine
1. Working theory:
This “V” face instrument adopts the modern aesthetic idea of v-shape, which is designed with a new combination technology. Combined with the 5MHZ bipolar & four polar RF and 650nm near infrared light technologies,the energy work on the deep tissue to dissolve the fatness strongly. Meanwhile it accelerates the metabolism, which helps the excretion for the dissolved fatness, to get the good effect of face sliming and sculpture.
1 V-01 Bi-polar RF
5MHZ high frequency Bipolar RF , the current flows only through the tissue which is between the two electrodes on the handpiece, to penetrate deep into the skin and affect the deeper dermis and subcutaneous layers, causing tightening and improvements to the underlying tissue structure, but with little change in skin texture or fine lines and wrinkling.
2 V-02 Four-Polar RF
Four-Polar’s key advantage is that it utilises 4 electrodes to allows for greater focusing of the RF energy and a broader RF wave density which means that it can achieve variable depths of penetration into the target tissue being treated. Collagen in the tissue is heated and stimulated to create an immediate visible contraction and lifting effect, followed by remodelling with new and accelerated collagen production in the deeper layers over time
3 V-03 LED mask
With LED Red light and Infrared light therapy. It can encourages collagen and elastin regeneration by stimulation fibroblast activity in the skin. It also helps to increase blood and oxygen flow to the skin and capillaries, therefore increasing cellular metabolism and strengthening the capillary walls
Bipolar RF use for:
1) Before the operation, both the customers and the beauticians should remove all the metal decorations.
2) The customer’s hair root part must be protected with gel or towel during the facial treatment, to avoid the static friction between the probe and the hair, which may cause the hair burn.
3) It is advised to use products with higher oil to moist the skin.
4) During the operation, the RF probe must touch the skin firstly, then move the probe around and adjust the energy. The Energy should be adjusted slowly from low to high level until the customer feels comfortable.
1) The headpiece can not leave the customer’s skin, and can not stay in the same position. It must be moved around on the face during the whole operation process.
2) The headpiece must touch the face totally, not only small part of it keeps working.
3) For the eye, the operation way is advised to make circles.
Four polar RF use for:
1) After first step, apply more cream or oil on the skin to make it moisten
2) The Four-polar RF probe must touch the skin firstly, then move the probe around and adjust the energy. The Energy should be adjusted slowly from low to high level until the customer feels comfortable
3) The headpiece can not leave the customer’s skin, and can not stay in the same position. It must be moved around on the face during the whole operation process.
4) The headpiece must touch the face totally, not only small part of it keeps working.
5) This probe is for face lifting and sculpting use. please move the treatment head and lift the facial skin base on the facial lines.
1) Before using the mask, please clean your face and the Uniform light membrane
2) Applying a serum, essence or moisturizer on the face, Use eyes patch or cotton to cover the eyes, then connected the uniform light membrane to the Mask. Then put it to the face. (if using the silk facial mask or Nonwovens facial mask, no need the uniform light membrane )
3) use the straps to fasten the mask
4) Adjust intensity base on the customer’s needs, usually we will suggest 4-5level, Treatment time: 10 minutes.
5) After treatment, please clean the membrane with water, and wipe down the Mask with Alcohol Pad.
The LED MASK TREATMENT, WE will suggest the treatment time is 10-15 minutes a day; interval time is not less than 8 hours for the second treatment
Notice: The operation time is decided by professional beautician according to the different skin type of the customer. It is advised to do the operation 10-15 minutes for each probe.